- 120-280 Volt (v)
- Silver & Black
- Machinical Extensometer
- Steeliness Steel
- 5-150 Kilograms (kg)
MECHANICAL EXTENSOMETER MODEL: FXT-3
Mechanical Extensometer is an attachment to Universal /Tensile Testing Machines. It measures the elongation of a test piece on load for the set gauge length. The least count of measurement is 0.01 mm and maximum elongation measured is upto 3mm.This elongation measurement helps in finding out the proof stress at the required elongation percentage.
WORKING PRINCIPLE :
- The required gauge length (between 30 to 120mm) is to be set by adjusting the upper knife edges and the scales.
- Hold the test specimen in the upper and lower jaw of Tensile / Universal Testing Machine.
- Position the Extensometer on the specimen. Position Upper clamp to press upper knife on the specimen.
- The extensometer will now be fixed to the specimen by spring pressure. Set zero on both the dial gauges.
- Start loading the test specimen and take the reading of load on the machine at required elongation or the elongation at required load.
- For better accuracies mean of both dial gauge readings should be taken as elongation.
Gauge length adjustable from
Dia. 1 to 20 mm. round or flat upto 20x20 mm
- Telegraphic Transfer (T/T), Cheque
- 1 Per Day
- 1-2 Week
- Africa, Middle East, Western Europe, Eastern Europe, South America, North America, Central America, Australia, Asia
- All India
C-76 A, Jyoti Colony, Street No. 7, 100 Feet Road, Shahdara, Delhi, Delhi, 110032, India